- Visible at a distance with crisp, colorful LED bar on top of unit with flip screen
- Remote viewing through the web
- Stands up to harsh conditions
- Duty cycle and k-factor multiplier options
- Flexible uses throughout your plant
- Easy to choose the target gas
- Quick start-up for emergency response
- Easy to use with protective equipment or hazmat gear
- Data logging and information management for total confidence
- Easy maintenance and long-lasting battery
Chemcassette tape-based with advanced self monitoring optics design and K-factor multiplier options.
Duty cycle options to save paper.
Height: 33.6 cm; Width: 18.3 cm; Depth without handle: 16.3 cm; Depth with handle: 24.1 cm
Concrete: 5/16 in x 2 in vibration-resistant stud anchor for concrete (McMaster-Carr 94475A185 or equivalent), add 0.25 in. to length when mounting bracket with sun shield.
Wood: 5/16 in. x 2 in. flange head lag screw for wood (McMaster-Carr 95526A375 or equivalent), add 0.25 in. to length when mounting bracket with sun shield.
0°C to 40°C for most gases/applications
0-100% RH for unit (Sample RH limited per tape/calibration). Sample line requires additional hardware to remove moisture in high RH conditions where condensing may occur. The sample must be non-condensing. Dry conditions may require humidification.
Automatic flow control with bypass system, higher flow at inlet to reduce sample time (internal bypass system); sample up to 100 ft, check manual for specific gases.
|Local Alarms/ Status||
Visual: LEDs for alarm, normal condition, fault and external power.
Audible: User selectable: Off, Low ~75 dB at 1 m, Medium ~85 dB at 1 m, High >90 dB at 1
4 large buttons, 3.5 in. Color LCD TFT display, web server. Flip screen.
Rolling up to 3 months (15-22 sec. with no gas reading, 1-2 sec. when reading gas), Event history (1500 events – approx. 1 year).
|Maximum Inlet/ Outlet Pressure Differential||
The overall maximum load on the pump between the inlet and the exhaust should not exceed 10” WC (24.8 mbars or 2488 Pascales).
250V AC, 6 A maximum
2.0 or later.
|Indoor/ Outdoor Use||
Both (the power supply provided is indoor only)
-1,000 to 3,000 ft. above sea level: standard; 3,000 ft. to 6,000 ft. above sea level: requires adjustment by Honeywell Analytics.
|Ingress Protection Rating||
|External Switch or Circuit Breaker Requirement (Description & Location)||
Meet or exceed all local codes and regulations.
Mount with no obstructions within 4 in. (10 cm) of either side or within 2 in. (5 cm) above and below the detector
Universal Line powered (90-260 VAC 50/60 Hz) for battery charger
~1.9 A at 24 VDC +/- 10% (including battery-charging current).
6+ hours under typical conditions – acts as battery back-up in fixed applications. Approximately 70% of its original capacity after 300 full charge/discharge cycles.
Input: 90~264 VAC, 1.2A, 47-63 Hz.
Output: 24 VDC, 3.75 A max.
Minimum: 24; Maximum: 14. AWG for relay and 4-20 mA outputs.
Relays: Alarm 1, Alarm 2, Fault (user configurable for normally open/closed)
Ethernet (with Modbus TCP/IP and web server).
USB port (for memory stick configuration/data transfer).
Communications connector and optional communications cable: 60 V, 5 A maximum
|Phosphine (PH3)||3-3000 ppb|
|Diborane (B2H6)||5-1000 ppb|
|Silane (SiH4)||0.03 - 50 ppm|
|Germane (GeH4)||50-2000 ppb|
|Hydrogen Selenide (H2Se)||2-500 ppb|
|Hydrogen Sulphide (H2S)||0.001-9.999 ppm|
|Hydrogen Fluoride (HF)||0.02-20 ppm|
|Hydrogen Chloride (HCl)||0.02-20 ppm|
|Hydrogen Bromide (HBr)||0.02-10 ppm|
|Boron Trifluoride (BF3)||0.05-10 ppm|
|Nitric Acid (HNO3)||0.02-20 ppm|
|Sulfuric Acid (H2SO4)||5-750 ppb|
|Hydrogen Iodide (HI)||0.2-10 ppm|
|Chlorine (Cl2)||0.005-5 ppm|
|Chlorine (Cl2)||0.01-5 ppm|
|Fluorine (F2)||0.01-10 ppm|
|Nitrogen Dioxide (NO2)||0.03-10 ppm|
|Chlorine Dioxide (ClO2)||20-1000 ppb|
|Ammonia (NH3)||0.01-150 ppm|
|Dimethylamine (DMA, C2H7N)||0.5-50 ppm|
|Tetrakis (Dimethylamido) Titanium (TDMAT, C8H24N4Ti)||0.01-20 ppm|
|Trimethylamine (TMA, C3H9N)||0.03-50 ppm|
|Phosgene (COCl2)||2-2000 ppb|
|Ethylchloroformate (ECF, C3H5ClO2)||0.02-30 ppm|
|Methylchloroformate (MCF, C2H3ClO2)||0.03-30 ppm|
|Toluene Diisocyanate (TDI, C9H6N2O2)||0.5-200 ppb|
|Methylene Bisphenyl Isocyanate (MDI, C15H10N2O2)||0.5-200 ppb|
|Hexamethylene Diisocyanate (HDI, C8H12N2O2)||0.5-150 ppb|
|Hydrogenated Xylene Diisocyanate (H6XDI, C10H14N2O2)||0.5-150 ppb|
|Methylene bis-(4- cyclohexylisocyante) (HMDI, C15H22N2 O2)||0.5-100 ppb|
|Isophorone Diisocyanate (IPDI, C12H18N2 O2)||0.5-150 ppb|
|Xylene Diisocyanate (XDI, C10H8 N2 O2)||0.5-200 ppb|
|Hydrazine (N2H4)||3-1000 ppb|
|Monomethyl Hydrazine (MMH, CH6N2)||3-2000 ppb|
|Dimethyl Hydrazine (UDMH, C2H8N2)||3-2000 ppb|
|Hydrogen Cyanide (HCN)||0.2-30 ppm|
|Sulphur Dioxide (SO2)||10-2500 ppb|
|Ozone (O3)||20-1000 ppb|
|Hydrogen Peroxide (H2O2)||0.1-3 ppm|